28 August 2020
- Last edited 28 August 2020
I also have an issue with wrong salt readings with Aqua Rite, I am using a T-15 cell, and it has the Check Salt and the Inspect Cell lights both flashing.
The reading is about 2600, on the display and when I press the diagnostic bottom 5 times to adjust the reading is -0.
I have also a TDS/EC COM-80 that I use to do my home testing, the reading on it is 3265 ppm.
To make sure everything is OK after those confusing readings, I took a water sample to the local pools supply to test, and the result was the condition of the water was ok, only the C Acid a bit low at 40; and to my surprise the salt was high, 3600 ppm, now the light of High Salt never went ON, on the panel.
Now before a replace some water I would like why the readings are so different s, is the the unit defective?
Forgot to add, that the cell was tested about a month ago, it was OK, and was cleaned 10 days ago.
I ll appreciate any help I can get to solve this issue.